have a commitment to offer only top-quality products and service to our customers.
This is the one area of our business where we feel that no compromise can be allowed. Of course, this means that we work together only with suppliers that can help us realize our quality vision.
Although we fulfill all of the ISO-stipulated requirements listed above, we go above and beyond whenever possible. Specifically, this means that our cartridge products have technical properties that match or exceed the properties you’d find from the original equipment manufacturer.
Force required to extract the bur from the chuck >30N
Force required to open the chuck mechanism using the push button 25N
The hardness of all parts including axis, on which the bearings are mounted 60
- 65 HRC
The radial deviation of a bur, from completely concentric, when turned measured 6cm from the end of the rotor‘s axis part < 0.015mm
The amount of rotor imbalance , expressed as a mass at a certain distance from the center of rotation < 0.2mgmm
In order to fulfill the vision spelled out in our quality philosophy, we have to ensure that we deliver the top technical characteristics for all products. In the case of our rotor assemblies, this means that absolutely all requirements of globally recognized standards are respected. Our rotors fulfill the requirements for the following ISO norms:
ISO 7785-1 & 2
Technical Requirements for air- and motor-driven dental handpieces.
Specifications for the burs and mandrels that are used with dental handpieces.
Description of the graphical symbols to be used for dental equipment.
Process description for sterilization of medical devices.
The draft new norm of dental hand pieces and motors, which replaces ISO 7785
and ISO 13294